EDPM Gauntlets & Sleeves
- Manufactured from EPDM, Ethylene-Propylene-Diene-Monomer.
- Good chemical resistance.
- FDA compliant.
- UV Resistant.
FEATURES & TECHNOLOGY
- Chemical Resistant
- Smooth Texture
- Great Comfort
Sleeves are supplied fitted with cuff and O-rings as standard.
For glove assemblies please follow this link.
£430.00 – £445.00
Download the technical specification of these EDPM Gauntlets and Sleeves using the following button. Alternatively, read the information on the tabs below.
Port Sizes Diameter: 203mm – 8″, 254mm-10”
Hand Sizes Available: 8 – 9 – 10
Thickness: 0.4mm – 0.015”
Length: 800mm – 31.5”
Recommended in-use lifespan: 1 – 2 Years (Light Duty)
Recommended Temperature Range: –20°C to + 130°C (4F to 266F)
- EDPM sleeves come fitted with Cuff and O-rings as standard.
- Glove Assemblies are also available.
FDA compliant EDPM Gauntlets and Sleeves. Antistatic, UV resistant and comfortable to wear, providing good tactile sensitivity.
These Laboratory Gauntlets provide good chemical resistance, very good steam sterilizability and also offer great durability.
Protection against microorganisms – According to DIN EN ISO 374-5: 2016.
Glove to protect against bacteria, fungi and viruses. The resistance against penetration was tested under laboratory conditions and only refers to the tested samples.
- Storage in original packaging. Dry conditions between +5°C and +25°C (41°F to 77°F)
- Keep away from direct sunlight; ozone sources or naked flame.
- Shelf life of a maximum of 3 years
Non Recyclable. Classified as non-toxic waste providing they have not been contaminated during use.
Some gloves may contain ingredients which are known to be a possible cause of allergies in sensitive persons who may develop irritant and/or allergic contact reactions. If an allergic reaction should occur seek medical advice immediately. This model does not contain any substances at levels that are known to, or suspected to, adversely affect user hygiene or health.
If you would like more information or advice on a product, please complete the following form.